Frequently Asked Questions
For enquiries about technical documentation please contact the Customer Care office at the following adress: email@example.com
For a detailed answer please use the Search engine or choose a category.
Most read F.A.Q.
Manually calculate PID control parameters
See PID calculation parameters manual
600-xxxx-0/1 Temperature Controller
Because you're working in EASY CONFIGURATION. To extend the functions, add + 128 to the value set in software protection.
Indicator 4/40T series
NO, because the MK1/MK3 models need 100/70mA , while the maximum courrent supplied by 4/40T is 50mA/24V.
AUTO-TUNING is an automatic procedure that calculates the best control parameters. The instrument runs this procedure when the parameter s.tun is enabled and when the variable is near the setpoint. How to activate auto-tuning. Example Controller 1600P-1800P.
The 4...20mA can be configured by jumper "S1" on the output "OUT 3" (see the user's manual) and setting +64/+65 on "rL.3" parameter in the "OUT" menu. S1 ON = 10V (closed by default) S1 OFF = 20mA (open)