Frequently Asked Questions
For enquiries about technical documentation please contact the Customer Care office at the following adress: email@example.com
For a detailed answer please use the Search engine or choose a category.
Most read F.A.Q.
40T-72: menu INP
It doesn't appear in the position described (after TP), but rather after HS. Unfortunately, there is an error in the user manuals.
Auxiliary power supply
Yes, but you have to install hardware jumpers and set a few parameters. You can find the instructions in the multilingual manuals at the heading APPENDICES. Given the diversity of the printed circuits, a complete explanation cannot be provided here. We advise you to contact CUSTOMER CARE.
Yes, this feature is available on model "2300", setting "IN.TYP" parameter +32. The same feature is available on the new "40T-48" or "600", enabling the virtual instrument condition in "PROT" parameter (+64) and setting the "S.I,S.o,S.U"...
There are usually two levels of protection: one hardware (jumpers) and one software (pro, prot). If these aren't correct, entry in the various configuration menus is denied. In particular, on new-generation instruments such as the 4-40-400-600-800-1600-1800, you have to set code 99 in PA/PASS to ...
IK1/MK1: reverse output
No. In the isolated version, the 20...4mA output must be specified in the order, and to change it you have to work on the hardware.